|Place of Origin:||China|
|Certification:||Calibration certificate(cost additional)|
|Minimum Order Quantity:||1 set|
|Delivery Time:||30 days|
|Payment Terms:||L/C, T/T|
|Supply Ability:||10 sets per month|
|Control Operation Mode:||PLC Control, 7 Inch Color Touch Screen Operation, With Infrared Remote Control||Glowing Filament:||Ф4mm ± 0.04mm (> 77 % Ni/20 ± 1 % Cr) Special Standard Ring Shape，horizontal Standing|
|Specimen Pressure On Glowing Filament:||0.95N±0.1N||Maximum Depth Of Heating:||7mm ± 0.5mm|
|Specimen Movement Speed:||10mm /s ～ 25mm /s||Testing Time:||30s±1s|
audio measurement equipment,
drop test equipment
IEC60065 Glow-wire Tester GW-2
Conforms to IEC60695-2-10-13, IEC60335-1, IEC60598-1, IEC60745,IEC60065 and etc.
This device simulates the thermal stress test of glowing component or heat source like overload resistance or ignition source which is caused in a short time. It is suitable for electric and electronic products, household appliances and other material to do fire hazard testing. It is also used in the No flame ignition source fire test to test initiation temperature and flammability index of related glowing filament.
Electrical equipment and its components and parts, solid electrical insulating materials and other solid combustible materials.
Test principle: the glowing filament whose material and shape is required(Ni77/Cr20) is heated to the testing temperature of 550 ℃ ～ 960 ℃ for 1 min by using the glow wire tester with large current. Then using required pressure (0.95N) vertically burns the test sample for 30s. By observing whether the test sample and bedding object ignite or the burning time, the users can judge the ignition danger of electric and electronic products. It is used for the ignition test, ignition temperature (GWIT) test, flammability test and flammability index (GWFI) test of solid insulating material and other solid combustible material.
|Control operation mode||PLC control, 7 inch color touch screen operation, with infrared remote control|
|Glowing filament||Ф4mm ± 0.04mm (> 77 % Ni/20 ± 1 % Cr) special standard ring shape,horizontal standing|
|Specimen pressure on glowing filament||0.95N±0.1N|
|Maximum depth of heating||7mm ± 0.5mm|
|Specimen movement speed||10mm /s ～ 25mm /s|
|Temperature calibration||Calibration temperature:960°±10°C, the hot wire by electric heating|
|Thermocouple: RS imported 1mm armour nickel-chromium/nickel/ nickel aluminum K-Thermocouple, temperature resistance 1100°|
|Silver foil: purity 99.8%, area :2m², thickness:0.06mm|
|Tray||A layer of silk paper (12~30g/m³) is covered on the 10mm thick pine board, with a distance of 200±5mm below the test sample|
|Test chamber||>0.5 cube,sample distance from the inside of the box is greater than 100mm, black yogon background,background illuminance ≤ 20Lx,with exhaust fan|
|Test procedure||The test sample is driven by the car and automatic run, the car will return automatically after glow-wire entered the test sample of 7mm|
|Lighting||Equip with lighting in the test chamber, with control|
Contact Person: Tessa Huang