|Place of Origin:
|calibration certificate (cost additional)
|Minimum Order Quantity:
|10 set /month
iec testing equipment,
laboratory test equipment
IEC60598 50mm Steel Ball Vertical And Horizontal Impact Electronic Control Device For Housing Equipment
The device is based on IEC60598-1 Luminaires –Part 1: General requirements and tests clause 4.13.4a), IEC60950-1 Part 1:
General requirements clause 4.2.5, IEC60065-1 Audio, video and similar electronic apparatus –Safety requirements clause 12.1.4 fig.8, 22.214.171.124 and other standard design and manufacture of special test equipment, suitable for housing equipment for anti-drop impact performance test, the device may vertical drop impact and the impact of swing level in two forms, a measurement accuracy, rational layout, easy operation, institutions and major manufacturers of laboratory detection of the ideal instrument.
1.The impact of the ball: Φ50mm belt loops, 500 ± 25g,
2.Swing center distance: 0-1.5 m (adjusted by adjusting the length of the rope);
3. Drop height: 0.5-1.5m adjustable
4.Lifting way: Electric
5.The clamping release mode: Electronic Control
6.According to the standard reference to FIG.
|Impact ball diameter
|Impact ball weight
|500 ± 25g
|Swing center distance
|The clamping release mode
1. Test preparation:
1.1 The sample is fixed in normal use position, ensure the support surface of the rigid support surface of choice for floors and walls do support surface.
1.2 The device connected to 220V power supply, according to the sample selection drop: Vertical drop or swing impact.
Vertical drop test: that is falling straight down, as shown, the rope end fixed clamping, vertical ball just touched the level of the sample surface (factory default height 1500mm), 1300mm drop test carried out for just control scale drop 200mm, the drop height 1300mm, then the ball will be placed in a vertical position release magnet card. Fixed wheel brakes, means for preventing movement.
Swing impact test: The wire rope fixed at one end, the ball will be placed at the level of release magnet card, then move the hook seat, make rope Bengzhi locking hooks seat. Then the ball is in a free state, just contact test a vertical surface (by remote control the height of the point of impact), the ball will again be placed in a horizontal position release magnet card. Fixed wheel brakes, means for preventing movement.
If you want to swing outward, you can remove ①②from the end of the rod, change it in one direction, then move it in, the rope end fixed clamping, the ball will be placed at the level of release magnet card, then move the hook seat, make rope straighten out and locking hooks seat, the same method as the swing inside .
2. Test operation:
Preparation work is done, press on the remote control to release the ball cross / vertical ball release button, proceed ball swing / vertical impact test. After the test, according to standards determined whether the sample qualified.
Contact Person: Tessa Huang