We provide you with industry leading test equipment,
advanced online system and solutions–anywhere in the world.
|Place of Origin:||China|
|Certification:||Third part calibration certificate(cost additional)|
|Minimum Order Quantity:||1 set|
|Packaging Details:||Aluminum Box|
|Delivery Time:||3 Days|
|Supply Ability:||100 sets/ month|
|Product Name:||Accessibility Wedge/Probe||Warranty:||1 Year|
|Standards:||IEC 62368-1, UL60950-1 And Etc.||Material:||Stainless Steel|
|Usage:||Testing Document Shredders||Probe Tip Width:||50mm|
|Probe Tip Thickness:||2mm||Lead Time:||3 Days|
Figure NAF2-3 Accessibility Wedge Probe,
UL60950-1 Accessibility Wedge Probe,
Thickness 2mm Accessibility Wedge Probe
UL60950-1 Figure NAF2-3 Accessibility Wedge, Probe Tip Thickness 2mm
Standard description(UL60950-1 clause F.4.4):
For HOUSEHOLD / HOME OFFICE DOCUMENT SHREDDERS , a warning statement shall not be used in lieu of construction features that prevent access to hazardous moving parts. Test Finger Probe
For HOUSEHOLD / HOME OFFICE DOCUMENT SHREDDERS , the Following additional compliance criteria apply:
The accessibility probe/wedge illustrated in figures NAF.2 and NAF.3 shall be inserted into each opening in the MECHANICAL ENCLOSURE . A force not exceeding 45 N for strip-cut type HOUSEHOLD / HOME OFFICE DOCUMENT SHREDDERS and not exceeding 90 N for cross-cut
type HOUSEHOLD / HOME OFFICE DOCUMENT SHREDDERS shall be applied to the probe/wedge in any direction relative to the opening. Before application of the accessibility probe/wedge, any MECHANICAL ENCLOSURES or guards that are removable without the use of a tool shall be removed. The probe/wedge shall not contact hazardous moving parts, including shredding rollers/mechanisms.
NOTE It is permissible for the mass of the accessibility probe/wedge to be factored into the overall applied force specified above.
Standard: IEC 62368-1 Figure V.4, UL60950-1 figure NAF.2 and NAF.3 and etc. Test Finger Probe
Application: It is used to test whether the hazardous parts of shredder can be touched by people.
Test sample: Paper shredder
Feature: It is manufactured according to the standard figure.
Parameters: Test Finger Probe
NOTE The thickness of the probe varies linearly, with slope changes at the following points along the probe:
Distance from probe tip
Picture for reference:
Contact Person: sales