We provide you with industry leading test equipment,
advanced online system and solutions–anywhere in the world.
|Place of Origin:||China|
|Certification:||Calibration certificate(cost additional)|
|Minimum Order Quantity:||1 set|
|Packaging Details:||Aluminum box|
|Delivery Time:||3 days|
|Supply Ability:||30 sets per month|
|Color:||White||Material:||Plastic + Stainless Steel|
|Warranty:||1 Year||Application:||Electronic Safety Tester|
|Baffle Plate Diameter:||75mm||Finger Length:||80mm|
Figure 8.4 Articulate Test Probe,
UL1278 Articulate Test Probe,
12mm Articulate Test Probe
UL1278 Figure 8.4 Articulate Probe Test Probe B - Verify Accessibility of Live Parts
Product information: An electrical part of a heater shall be so located or enclosed that protection against unintentional contact with uninsulated live parts will be provided.
8.4 To reduce the likelihood of unintentional contact that may involve a risk of electric shock from an uninsulated live part or film-coated wire, an opening in an enclosure shall comply with either (a) or (b).
a) For an opening that has a minor dimension less than 1 inch (25.4 mm), such a part or wire shall not be contacted by the probe illustrated in Figure 8.3.
b) For an opening that has a minor dimension of 1 inch or more, such a part or wire shall be spaced from the opening as specified in Table 8.1.
Exception: An integral enclosure of a motor need not comply with these requirements if it complies with the requirements in 8.5.
With respect to a part or wire as mentioned in above, in an integral enclosure of a motor as mentioned in the exception to above:
In a directly accessible motor , an uninsulated live part cannot be contacted by the probe illustrated in Figure 8.4
Figure 8.4 probe shall be applied to any depth that the opening will permit; and shall be rotated or angled before, during, and after insertion through the opening to any position that is necessary to examine the enclosure. The probes illustrated in Figure 8.4 shall be applied in any possible configuration; and, if necessary, the configuration shall be changed after insertion through the opening.
Standard: IEC 61032 figure 2, IEC60950 figure 2A, IEC60884, IEC60335, UL507, and UL1278 figure 8.4.
Application: This probe is intended to verify the basic protection against access to hazardous parts. It is also used to verify the protection against access with a finger.
Test sample: Easily accessible enclosure live parts or mechanical parts.
Feature: This test finger probe consists of dito, finger, base and insulated handle which simulates the characteristics of the human hand. It has two movable joints, which can be curved at 90°. Customizable: it can be used for anti-electric shock test when equipped with pluggable banana plugs and amphenol connector at the end of the handle , or used for enclosure protection test when open threaded hole of M6 at the end of the handle (connected with pull and push dynamometer).
|Name||Standard Test Finger|
|Fingertip to baffle||180±0.2|
|Fingertip cutting bevel angle||37°|
|Test finger diameter||Ф12|
|A-A Section diameter||Ф50|
|A-A Section width||20±0.2|
IP1~4 TEST PROBE KIT
Contact Person: sales