|Place of Origin:||China|
|Certification:||calibration certificate (cost additional)|
|Minimum Order Quantity:||1 set|
|Delivery Time:||25 days|
|Supply Ability:||10 set /month|
|Pendulum Rod Angle:||0-90 ° Adjustable||Angle Display:||Dial Scale|
|Pendulum Rod Length:||1000mm||Striking Elements:||2J,5J ,10J,20J, 50J|
|Height Of Fall:||200,300,400,500mm||Rod Diameter For 2J:||9mm|
|Rod Diameter For 5-50J:||15.9mm||Wall Thickness Of Rod:||0.5mm|
50J Impact Testing Machine,
IEC60068-2-75 Impact Testing Machine,
1000mm Arm Pendulum Hammer Test Apparatus
IEC60068-2-75 up to 50J Impact Testing Machine-Pendulum Hammer Test Apparatus With 1000mm Rigid Arm
IEC60068-2-75 Clause 4.2:
The test apparatus consists basically of a pendulum rotating at its upper end in such a way as to be kept in a vertical plane. The axis of the pivot is at 1 000 mm above the measuring point. Impact Testing Machine
The pendulum is composed of a nominally rigid arm and of a striking element complying with the requirements of table 1 .
For testing heavy, voluminous or difficult to handle specimens, a portable pendulum may be used. It shall comply with the above description but its pivot may be fixed directly on the specimen or on a movable structure. In this case, it shall be ensured that, before the tests, the axis of the pendulum is horizontal, that its fixing is sufficiently rigid, and that the impact point is in the vertical plane passing through the axis.
In all cases, when the pendulum is released, it shall be allowed to fall only under the influence of gravitational force.
Impact Testing Machine
Standard: IEC60068-2-75 Environmental testing – Part 2: Tests – Test Eh: Hammer tests, IEC60068-2-63, IEC62262 Degrees of protection provided by enclosures for electrical equipment against external mechanical impacts (IK code), IEC60439-5 Low-voltage switchgear and controlgear assemblies - Part 5:Particular requirements for assemblies for power distribution in public networks
Application: This device is suitable for mechanical impact strength test to electric cabinets and other sheet metal, shell of Non metal electric energy metering cabinet, substation cable switchboard and cable junction box Impact Testing Machine
Test sample: The shell of electric energy metering cabinet, substation cable switchboard and cable junction box and similar equipment
Feature: Equipment using the car structure, can be moved left and right, forward and backward, the upper and lower impact point height adjustable, and the impact angle 0-90°adjustable. The device is a high energy pendulum impact device for 2J and above.
|Pendulum||Length: 1000mm; external diameter: 9mm; steel pipe wall thickness: 0.5mm. Composition of the equivalent mass with impact element|
|Impact point||Up and down adjustable, stroke 200-1200mm, height can be customized|
|Impact element||2,5,10,20,50J impact hammer|
|Drop height||Scale display Impact Testing Machine|
Impact energy, equivalent mass and drop height table
|Equivalent mass /kg||0.5||1.7||5||5||10|
|Drop height ±1%mm||400||300||200||400||500|
Notes: 1. See note in IEC 60068-2-75 3.2.2
Contact Person: Tessa Huang