We provide you with industry leading test equipment,
advanced online system and solutions–anywhere in the world.
|Place of Origin:||China|
|Certification:||Calibrate Certificate (Cost Additional )|
|Minimum Order Quantity:||1 Set|
|Delivery Time:||30 days|
|Supply Ability:||10 set /month|
|Turntable Diameter:||Ø600mm Customized||Sample Holder:||Ø600mm Customized|
|Turntable Speed:||1r/min-10r/min, Adjustable||Tiltable Angle:||15°|
|Test Angle Of Rotation:||Continuous Rotation||Load Bearing:||150kg|
IEC60529 Tiltable Rotating 600mm Stage With Independent Control Box 150KG Adjustable Height
The rotating table is manufactured in accordance with the standard of Clause IEC60529 testing purpose.The loading platform can be inclined 15°to meet with IPX2 inclined testing requirement.
1. Turntable installation
Loosen the screw on the upper part of the supporting column, rotate the upper support stage, can adjust the height of turntable.
The bottom is provided with four supporting feet, and the supporting feet is adjusted to make the turntable perpendicular to the ground. Two screws were installed opposite at the bottom, loosen the two screws, can tilt the turntable. Tightening the screws according to the screw hole position after tilt, the support stage with an angle of 15 degrees to vertical turntable.
Place the sample to be tested on the turntable, adjust the tilting screw at the bottom of the turntable as needed, and tilt the turntable 15°. Press the start button and adjust the speed regulator according to the desired speed. If the sample is too heavy, the upper support stage can be removed and placed directly on the underside aluminum plate.
|Turntable speed||1r/min-10r/min, adjustable|
|Test angle of rotation||continuous rotation|
|Adjustable height of the adjustable telescopic tube||1200mm-1800mm|
|Load bearing||≤150kg(Underside aluminum plate)|
Control Function Overview
|POWER||Used for turning on /off the power supply, when the power supply is turned on, the stop indicator is bright.|
|START||Press this button, the turntable starts to rotate, and the green indicator lights up.|
|STOP||Press this button, the turntable stop to rotate.|
|SPEED REGULATION||Press the start button and adjust the speed control knob to achieve the required test speed|
Common Faults And Treatment
1. The power-on indicator does not light, check if the fuse and indicator are burnt out and replace it
2. The turntable cannot be turned, check if the speed control light is on.
Note: This equipment is powered by high voltage and must be protected against electric shock during use. The operator must strictly abide by the operating procedures and the outer casing of the equipment and the metal casing touched by personnel must be well grounded.