We provide you with industry leading test equipment,
advanced online system and solutions–anywhere in the world.
|Place of Origin:||China|
|Certification:||Calibrate Certificate (Cost Additional )|
|Minimum Order Quantity:||1 set|
|Delivery Time:||30 days|
|Supply Ability:||10 set /month|
|Adjustable Height:||300mm, 1200mm-1500mm||Turntable Diameter:||Ø600mm (customized)|
|Turntable Speed:||Maximum 1r/min, Adjustable||Tiltable Angle:||15°|
|Test Angle Of Rotation:||Continuous Rotation||Load Bearing:||150kg|
Rotating Stage Ingress Protection Test Equipment,
1r/Min Ingress Protection Test Equipment,
IPX6 Ingress Protection Test Equipment
15 Degrees Tiltable Rotating Stage Ingress Protection Test Equipment 1r/min For IPX1-IPX6
1 4.2.2 Test for second characteristic numeral 2 with the drip box
The dripping device is the same as specified in 1 4.2.1 adjusted to provide the water flow rate specified in table 8.
The table on which the enclosure is placed does not turn as in the case of the test for the second characteristic numeral 1 .
The enclosure is tested for 2,5 min in each of four fixed positions of tilt.
These positions are 1 5° on either side of the vertical in two mutually perpendicular planes (see figure 3b)).
The total duration of the test is 10 min. Ingress Protection Test Equipment
The rotating table is manufactured in accordance with the standard of Clause IEC60529 testing purpose.The loading platform can be inclined 15°to meet with IPX2 inclined testing requirement. It can be used with IPX1 ~ IPX6 test.
1. Turntable installation Ingress Protection Test Equipment
Loosen the screw on the upper part of the supporting column, rotate the upper support stage, can adjust the height of turntable.
The bottom is provided with four supporting feet, and the supporting feet is adjusted to make the turntable perpendicular to the ground. Two screws were installed opposite at the bottom, loosen the two screws, can tilt the turntable. Tightening the screws according to the screw hole position after tilt, the support stage with an angle of 15 degrees to vertical turntable.
2.Start test Ingress Protection Test Equipment
Place the sample to be tested on the turntable, adjust the tilting screw at the bottom of the turntable as needed, and tilt the turntable 15°. Press the start button and adjust the speed regulator according to the desired speed. If the sample is too heavy, the upper support stage can be removed and placed directly on the underside aluminum plate.
|Sample holder||Ø600mm Ingress Protection Test Equipment|
|Turntable speed||1r/min, adjustable|
|Test angle of rotation||continuous rotation|
|Adjustable height of the adjustable telescopic tube||
|Load bearing||≤150kg(Underside aluminum plate)|
Control Function Overview
|POWER||Used for turning on /off the power supply.|
|CLOCKWISE||Press this button, the turntable starts to clockwise rotate, and the indicator lights up. Ingress Protection Test Equipment|
|ANTICLOCKWISE||Press this button, the turntable starts to anticlockwise rotate, and the indicator lights up.|
|SPEED REGULATION||Press the start button and adjust the speed control knob to achieve the required test speed|
Contact Person: sales