We provide you with industry leading test equipment,
advanced online system and solutions–anywhere in the world.
|Place of Origin:||China|
|Certification:||Calibration Certificate(cost additional)|
|Model Number:||CHA - 1|
|Minimum Order Quantity:||1|
|Delivery Time:||30 days|
|Payment Terms:||L/C, T/T|
|Power Supply:||AC230V 50Hz||Crank Speed:||30r / Min ( Can Be Preset )|
|Horizontal Moving Distance Of The Hose:||300mm||Working Station:||3 Stations , Synchronous Working|
Current - Carrying Hoses Brasion Resistance Electrical Appliance Tester 30R / Min
Standard : This device meets the standard of IEC60335-2-2, clause 21.102 figure 101. Suitable for the Current-carrying hoses abrasion resistance test.
Technical Parameters :
|Power supply||AC230V 50Hz.|
|Control and operation||PLC intelligent electrical control, touch screen operation|
|Working station||3 stations, synchronous working|
|Crank speed||30r/min, can be preset|
|Horizontal moving distance of the hose||300mm|
|Belt of abrasive cloth moving speed||0.1m/min|
|Test times||0-999999 times, can be preset|
|Abrasive belt||Corundum grit size P 100, as specified in ISO 6344-2.|
|Mass weight||1 kg|
|Mass has a maximum distance from the centre of the roller||600 mm|
|Size and weight||1000×900×1100mm, 100kg|
One end of the hose is attached to the connecting rod of the crank mechanism shown in Figure 101. The crank rotates at 30 revolutions per minute resulting in the end of the hose moving horizontally backwards and forwards over a distance of 300mm. The hose is supported by a rotating smooth roller over which a belt of abrasive cloth moves at a speed of 0.1 m/min. The abrasive is corundum grit size P 100, as specified in ISO 6344-2. A mass of 1 kg is suspended from the other end of the hose, which is guided to avoid rotation. In the lowest position, the mass has a maximum distance of 600 mm from the centre of the roller. Test test is carried out for 100 revolutions of the crank.
After the test, basic insulation shall not be exposed and the electric strength test of clause 16.3 is carried out between the conductors connected together and the saline solution.
Picture for Reference :