We provide you with industry leading test equipment,
advanced online system and solutions–anywhere in the world.
|Place of Origin:||China|
|Certification:||Calibration certificate(cost additional)|
|Minimum Order Quantity:||1 Set|
|Delivery Time:||30 days|
|Payment Terms:||L/C, T/T|
|Supply Ability:||30 set per month|
|Test Cycle:||192 Cycles||Test Current:||AC 1-99.9A|
|Voltage Drop Range:||0-199.9mv||Power Supply:||Customizable|
electric socket tester,
electrical plug tester
IEC60884-1 Electrical And Thermal Stresses Test Device For Screwless Terminals Stepless Adjust Load Current
IEC 60884-1 'Plugs and socket-outlets for household and similar purposes – Part 1: General requirements' clause 12.3.11.
Compliance is checked by the following tests a) and b), which are carried out on five screwless terminals of socket-outlets which have not been used for any other test.
Both tests are carried out with new copper conductors.
a) The test is carried out loading the screwless terminals for 1 h with an alternating current as specified in table 10 and connecting rigid solid conductors 1 m long having the nominal cross-sectional area as specified in the same table.
The test is carried out on each clamping unit.
Table 10 – Test current for the verification of electrical and thermal stresses in normal use for screwless terminals
Nominal cross-sectional area
of the conductor
During the test the current is not passed through the socket-outlet, but only through the terminals.
Immediately after this period, the voltage drop across each screwless terminal is measured with rated current flowing.
In no case shall the voltage drop exceed 15 mV.
The measurements are made across each screwless terminal and as near as possible to the place of contact.
If the back connection of the terminal is not accessible, the specimens may be adequately prepared by the manufacturer; care shall be taken not to affect the behaviour of the terminals.
Care shall be taken to ensure that, during the period of the test, including the measurements, the conductors and the measurement devices are not moved noticeably.
b) The screwless terminals already subjected to the determination of the voltage drop specified in the previous test a) are tested as follows.
During the test, a current equal to the test current value given in table 10 is passed.
The whole test arrangement, including the conductors, shall not be moved until the measurements of the voltage drop have been completed.
The terminals are subjected to 192 temperature cycles, each cycle having a duration of approximately 1 h and carried out as follows:
– the current flows for approximately 30 min;
– for a further period of approximately 30 min no current flows.
The voltage drop in each screwless terminal is determined as prescribed for the test of a) after every 24 temperature cycles and after the 192 temperature cycles have been completed.
In no case shall the voltage drop exceed 22,5 mV or twice the value measured after the 24th cycle, whichever is the smaller.
After this test an inspection by normal or corrected vision without additional magnification shall show no changes evidently impairing further use such as cracks, deformations or the like.
In addition, the mechanical strength test according to 12.3.10 is repeated and all specimens shall withstand this test.
It is use for test the screwless terminals shall withstand the electrical and thermal stresses occurring in normal use.
|Parameters Name||Parameter Data|
|Input supply||AC120V±10%, 60Hz (Customizable)|
|Test cycles||192 cycles, current flow for 30min, no current flow for 30min|
|Test current||AC 1~99.9A±(3%±1A)|
|Voltage drop measurement||0~199.9±(3%±1mv)|
Contact Person: sales